Core Lab is pleased to welcome Sanchez Technologies to the Core Lab Family
The Core Lab Instruments (CLI) ElectroMagnetic Viscometer (EMV) System is based
on the Cambridge Viscosity SPL440 sensor and the VP2000 viscometer controller. The
SPL440 sensor is installed in a dedicated oven, permitting operation from just below
ambient (with air vortex or lCO2 cooling) to its maximum operating temperature of
190 °C (374 °F). The cylindrical viscosity sensor is mounted in a cradle that allows
it to be orientated in four (4) fixed positions for loading, operating, and torque
opening/closure of the removable end of the sensor.
The oven for this EMV system has been enlarged to accommodate a Degasser Fitting on the top of the EMV cell in
place of the standard adapter. A system of three valves mounted inside the oven,
but operated externally; allow the sample to be charged to the sensor. An in-line
filter in the oven on the inlet side of the system removes particulates which could
interfere with the measurements. A pressure transmitter, mounted just outside the
oven, monitors the pressure.
The pressure transmitter is mounted above an external
bleed/purge valve which allows the dead-space up to the transmitter body to be cleaned
between analyses to minimize risk of sample carry-over. The pressure rating of the
system is dependent on the weakest link of the installed components. Three (3) pressure ranges are available,
10,000, 15,000, or 20,000 psi. The SPL440 sensor is rated to 20,000 psig.
The oven is mounted on the top of a modular, roll around aluminium framework which
also supports the VP2000 control box and the CLI built Control Box. The VP2000 control
box controls the piston movement inside the SPL440 as well as monitoring the internal
SPL440 sensor. It holds the calibration data for all the supplied sinkers and uses
these data to produce a raw viscosity figure. The raw viscosity figure is subject
to correction for pressure and temperature distortion effects on the piston and
The RS232 connection inside the VP2000 box is connected through
to the front panel of the CLI EMV Control Box to allow logging or setup operations
to be performed. This system also has an RS232 connector on the front panel to allow
communication with the pressure display. A LabVIEWTM pressure logger application
is provided on the DVD in the documentation folder.